Experience of using automatic image measuring instrument
Start up method of automatic image measuring instrument:
Turn on the device 2. Turn on the computer 3. Turn on the software
4.Machine return to zero 5. Normal operation
The optical film image measuring instrument is one of the more advanced precise and efficient measuring methods at present. Its working principle is that the measured workpiece (placed on the workbench) is illuminated by LED surface light or contour light (in the base), and then the image is captured by the zoom objective lens color CCD camera, and the computer is connected to aim and measure it through the cross line. The workbench drives the optical ruler to move in X and Y direction Move up the direction, and process by data card to complete the measurement and image system
The image window displays the dynamic image obtained by the camera.
The image window displays the dynamic image obtained by the camera. This window is the main window for image measurement.
In the coordinates (L: 0.7 -- m29.18) displayed in (1), if M has no digital display, it is abnormal,
Pixel correction is needed
Operation method: first click the button on the annotation toolbar, then click the element to be annotated in the image area or drawing area, move the mouse, and when the mouse pops up again, it means that the annotation is completed.